Get calibration certificates for your sensors.
All tests are performed against NIST traceable measurements and as per ISO 17025 standards.
Temperature: 3 reference points at 0°C, 20°C & 50°C.
Humidity: 3 points at 35, 50 & 70 %rH at 35°C.
IR Spot (EST Kit): 3 points at 33 & 35°C at 5 and 15cm distance.
Thermal Camera (EST Kit): 3 points at 33 & 35°C at 5 and 15cm distance.
Optional Hot Swap: we ship you calibrated sensors first; you then return your sensors.
Our in-house lab issues a unique calibration certificate per sensor it tests. We do strictly follow the ISO 17025 standards when testing your sensors.
We verify if your sensors are within tolerance for the different test points. Certificates include the calibration equipment used with serial numbers. The calculated uncertainty, calibration date and due date.
Like temperature sensors, annual calibration is recommended due to sensor drifting. When used in a dirty environment (air pollulants) or outdoors, then the frequency should be every 6 months for a calibration. With our hot swap option, we ship you a calibrated sensor before your return yours.
We check your sensor against 3 humidity points (35%, 50% and 70%).
Tests are performed using one or more calibrated and NIST traceable reference humidity sensors.
Learn more on sensor drifting